Buy ASTM D (R) TEST METHOD FOR SEDIMENT IN CRUDE OILS AND FUEL OILS BY THE EXTRACTION METHOD from SAI. Determines sediment content of crude oil and fuel oils by extraction with toluene. Find the most up-to-date version of ASTM D at Engineering
|Published (Last):||22 October 2004|
|PDF File Size:||2.36 Mb|
|ePub File Size:||20.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
Ramin’ USA Corporation
Fluctuating for the first two and then constant? Your bank will convert d73 payment done using credit or debit cards into EMI in working days. Hi guys perhaps this is a long shot. For payments done using credit and debit cards, to make this a No Cost EMI offer, the interest amount will be discounted from the price of your order.
See more product details. Write a product review. That is Sediment by extraction. It is agitated for homogeneity and dd473 decanted into a beaker, measuring around I have been following the ASTM method stated above. Andy on April 27, Purpose Determines sediment content of crude oil and fuel oils by extraction with toluene. What men are poets who can speak of Jupiter if he were like a man, but if he is an immense spinning sphere of methane and ammonia must be silent?
What I will do; I will run 3 or 4 times and see what I get again. Cookies help us provide you a better browsing experience. See questions and answers. How repeatable are those measurements? What happens to the weight of a conditioned thimble? The bank will charge GST on the interest amount.
Glass water cup Block-tin coil condenser, Erlenmeyer extraction flask, ml capacity, extraction thimble suspended by means of a corrosion resistant wire. You want get good results, until you have the initial thimble weight under control.
The problem I am having is getting the weight of the thimble steady.
Sediment Extraction Apparatus, Manufacturer, Supplier, India
The procedure is as follows – Condition thimble in test apparatus with fresh Toluene for 1hr commencing the satm point at which v473 Toluene fully passes through thimble – After 1 hour has passed place thimble in Laboratory oven at c for 1 hour – After 1 hour in oven has passed place thimble in beaker in desiccator for 1 hour – After 1 hour in desiccator weigh – Repeat above steps and aastm weight until consistent weight is achieved The problem is getting the weight of the thimble consistent and achieving a positive weight after passing HFO through it.
Product details Item model number: Be the first to review this item. Image Unavailable Image not available for Colour: Indeed the thimble has to weigh in consistently before the test can be accurately performed.
Powered by SMF 2. You can gave us a table of your results. Any help will be much appreciated.
From experience I wouldn’t expect any day to day deviation to be great enough to make a huge different in the result. The sample is then heated in the beaker above a hot place at c to evaporate any water in the sample.
I am extracting heavy fuel oils through astn thimble using the solvent Toluene. Thanks for the quick reply. I wasn’t getting consistency with the conditioning before but I am now using double asttm thimbles so will astmm with the new ones. Have one to sell? Inclusive of all taxes. Are you following it carefully? In the ASTM method they use a thimble ‘weighing not less than 15 g and not more than 17 g’ whereas mine weighs circa 4g straight out of the box.
Read This Before Posting. Obviously this should not happen, it should always be greater than. Share Facebook Twitter Pinterest. I just like to shoot straight. I am calculating the sediment weight to 4 asgm places so the thimble weight before testing must be representative.
Seta Extraction Apparatus | ASTM D | IP 53
Are you a business? Hey, I’m not judging. Maybe someone has experience with this test? Share your thoughts with other customers. In terms of the sample.
See if I can get consistency that way. I am generally very familiar with these tests and very rarely have problems but there is one I am totally new to. Audible Download Audio Books.